The EC help provides an easy access to the information related to the use of Keysight EC Network Analyzer. Pressing Help key on the front panel. EB/EB, please see the Programming Manual. • Installation and Quick Start Guide (Part Number: Ex1, attached to optional ABA). This manual . The EC complies with INSTALLATION CATEGORY II as well as The Service Manual is a guide to servicing the EC ENA Series Network Analyzer .
|Published (Last):||19 July 2009|
|PDF File Size:||11.36 Mb|
|ePub File Size:||20.1 Mb|
|Price:||Free* [*Free Regsitration Required]|
A message manyal appears after the completion of recovery process. Safety Safety Summary The following general safety precautions must e5071cc observed during all phases of operation, service, and repair of this instrument. The EC restarts automatically. Turns on the EC. The printing date changes when a new edition is printed. CN or above Step 1. Close all other applications you currently open.
Dangerous Procedure Warnings – Warnings, such as the example below, precede potentially dangerous procedures throughout this manual. For more information, see Modifying Calibration Kit Definition. Connect the keyboard to the EC.
To use this function, you must prepare for recovery in advance.
Atec Agilent-EC User Manual | 60 pages
Press Adjust Touch Screen. A maual bar appears Figure indicating the progress of the Recovery Image setup. The dialog box disappears automatically after a few seconds.
Inset the key in the lock and turn anti-clockwise. Click Done to terminate the 1-port calibration process.
Help File – User Manual for E5071C ENA Network Analyzer (Operation and Programming)
Click Open to start the calibration measurement. Click Load to start the calibration measurement. A total of twelve error terms, six each in the forward direction and the reverse direction, are used in a calibration. The HDD revision can be checked using the following procedure: Click Select Ports, then select the test ports on which you will perform full 2-port calibration. CN or below Returns the contents of the C and D drives to a user-specified state. This calibration effectively eliminates the directivity error, crosstalk, source match error, frequency response reflection tracking error, and frequency response transmission tracking error from the test setup in a transmission or reflection test using those ports.
Select the saved backup image file, then click Next button.
Also, never save the mnaual of the drive to the C, D or F drive. To avoid injuries, always disconnect the power and discharge circuits before touching them. EC firmware is executed, then the measurement display appears.
NOTE With no operation on the touch screen calibration screen for a preset time, it automatically closes and the previous measurement screen reappears.
Push the enter to select Yes to start the recovery of the factory backup.
Remove the two stickers item 1. After a while, the EC restart automatically.
Agilent Technologies EC Instruction manual |
Reconfiguring Touch Manial Settings Step 1. Shut down the EC. The System Properties dialog box appears. Touching the four locations described above with your finger automatically concludes the touch screen calibration.
To use this function, you must create the user backup image in advance. The error correction function will also be automatically enabled.