Keyboard Layout v Fluke B/1 96B-C/B-C Service Manual List of Figures For the mA. current D pin 3 is connected to pin 1, and pin 13 is. Manual de Servicio Fluke b – Ebook download as PDF File .pdf), Text File . txt) or trigger level mV) or (R the switches in D are all open. F HP Deskjet F HP Deskjet D HP Deskjet D HP Deskjet D . would work while others would not, and maybe it’s some strange fluke.
|Published (Last):||15 April 2008|
|PDF File Size:||20.44 Mb|
|ePub File Size:||3.34 Mb|
|Price:||Free* [*Free Regsitration Required]|
Check also Ml and Ml for the same voltage. The Fly Back Converter is off. These units can be used for the Fluke C and C only. Now press Sfe’S Continue to start the New version bottom cover Display Pixel Test Pattern 7. This IC takes care of a correct power state of D Verify that the backlight wires are d156 to minimize interference voltages. Select the Ohms or Volts function. Another internal current source supplies a current to R Observe the Input B reading and check to see if it is within the range shown under the appropriate column of table Gently remove the black or transparent protective cover; Step 2: By flike the pulses in low pass filters RCsoftware controlled DC voltages are generated.
The test tool is operative now. flke
If you proceeded to calibration step N for example step CLthen return to a previous step for example step CLand then calibrate this step, the complete rluke calibration becomes invalid; then you must repeat the calibration starting at 5. Power the test tool via the power adapter input using the BC power adapter.
HP C9352AE (22) Multi Colour Cartridge for Inkjet Printer
Set the A to operate OPR. Do the Final Calibration, section 5. From the measured voltage and supplied fluek the resistance value is calculated. This decreases the electromagnetic emission. If not OK see section 7. Continue through all calibration points of Table 8. Wait until the display shows the status: Remove all input connections from the test tool. The Input A and Input B direct path monitors the input signal.
The unit is not repairable. From November onwards, also the Fluke 1. Check the supply voltages V This ASIC includes a signal path and a trigger path. The 10 kHz bandwidth limiter rejects calibrator noise.
Full text of “FLUKE “
Get to Know Us. If none of the voltages is correct, then the fly back converter does not run correctly, continue at step 2.
Do the Contrast Calibration Adjustment, section 5. Most probably the unit will not meet the specifications if the actual calibration data are saved. Connect the test tool to the A as for the previous test see Figure Set the tilt stand to a 45 -degree position respective to the test tool bottom.
This prevents saturation and distortion in the HF transformer. The secondary transformer current is sensed by R, and fed back to N pin 7 and pin 4 for regulation vluke the PWM buck regulator output voltage.
The battery pack must be installed. V4, see step 2. This happens in particular at high temperatures. Search the history of over billion web pages on the Internet.